Photopatterning, module, software,
platform & reagents…
The PRIMO solution gives researchers complete freedom and full control over their manipulations to optimize the experimental conditions easily. PRIMO allows any shape of protein pattern to be generated rapidly, to precisely align or overlay several proteins (up to 3 without restrictions depending on the proteins used and their interactions) and to control the local density of protein adsorbed to the substrate.
The PLPP accelerates the degradation reaction of the anti-fouling polymer during photopatterning (a few seconds instead of a few hours). However, microfabrication manipulations with SU8 resins or NOA do not require the use of PLPP, since these products are photosensitive in the PRIMO wavelength (375 nm).
The total time to create a pattern (excluding the design step) ranges from 10 to 15 minutes. The illumination time varies and must be optimized for the patterns projected and the objective used. After each UV illumination and degradation step for the anti-fouling polymer (anti-adhesive), about 5 minutes must be allowed for the proteins to adsorb (time varies with the concentration placed on the substrate).
Photopatterning is a subtractive technology. It first aims to cover a substrate with an anti-fouling polymer that prevents molecules and cells from adhering. The PRIMO system is then used to degrade this anti-adhesive surface locally, under the combined action of UV and the photoactivatable reagent (PLPP).
We recommend using PLL-PEG. We regularly develop new protocols to enhance the stability of surface coatings.
We recommend coating on the same day. However, our R&D team regularly develops new protocols for specific substrates making it possible to keep coated slides for later use.
The resolution is 1.2 µm over the entire field of view of approximately 500x300 µm with a 20x objective.
PRIMO allows all desired images to be patterned, without size limitation.
PRIMO can be used to pattern all standard cell culture substrates: PDMS, glass, plastic, NOA, SU8. New substrates are regularly tested and validated by our in-house R&D team. In-house data available on request.
All photoresist sensitive at 375 nm can be used with PRIMO. Our R&D team mainly works with SU8 photoresists.
PRIMO can be used to pattern all surfaces, whether flat, curved or microstructured. For example, it is possible to pattern proteins on microwell walls or PDMS micropillars. In-house data available on request.
By using a 20x objective, the depth of field for the laser passing through a transparent substrate is 200 µm to 300 µm.
We recommend using the transfer technique by patterning proteins with PRIMO onto a slide that will be laid on the hydrogel then removed. The UV illumination phase to photopattern proteins onto hydrogels modifies their physical properties and therefore the initially-determined experimental conditions.
Our team and users employ more than ten different proteins, including Fibrinogen-488, Fibrinogen‐647, Fibronectin, GFP, Neutravidin‐488, Neutravidin‐647, PLL‐PEG‐Biotin, Protein A‐647 and Streptavidin, as well as primary and secondary antibodies.
Yes, experiments conducted with PRIMO can be carried out under sterile conditions, even if the PRIMO device is not set up in a room assigned to this purpose. In such cases, we recommend performing all rinsing and incubation steps in a biosafety cabinet (BSC), then putting the sample in a closed culture plate that can then be placed on the motorized stage of the PRIMO photo-patterning device.
The stability period of cell patterns depends on the substrate used and cell type cultured. Under certain conditions, the cells can adhere to the protein patterns for up to 2 weeks. Our R&D team regularly develops new protocols that are made available to our users.
The PRIMO unit contains a UV laser operating at 375 nm.
The power of the laser exiting the tube lens is between 6 mW and 8 mW. At objective outlet, the power is variable and depends on the objective. Using the Nikon S Plan Fluor ELWD 20x/ 0.45 reference objective, it is between 5 mW and 6 mW; the intensity is approximately 29 mW/mm².
The dimensions of the PRIMO unit are 46.2 x 32.5 x 7.2 cm (length x width x height). It weighs 15 kg and can be supported by a base supplied at installation.
The image file formats used to generate protein patterns must be in tif (pixels) or pdf (metric) format.
We recommend drawing patterns using the Inkscape® software (open source vector design software similar to Adobe Illustrator®), for which a user guide is available here. This software creates drawings in both pixels and millimeters and saves them in different formats recognized by the Leonardo software.
PHOTOPATTERNING WITH PRIMO: watch the recording of the webinar organized by Alvéole to present PRIMO before it was launched, how its photopatterning technology works and the range of applications it covers in cell biology.